To the Editor
In the setting of the coronavirus disease 2019 (COVID-19) pandemic, healthcare facilities have been forced to adopt strategies to extend or reuse personal protective equipment (PPE) such as N95 filtering facepiece respirators and surgical face masks.1 Cloth face masks worn in public settings where social distancing cannot be maintained are also typically reused multiple times between laundering. A variety of strategies for decontamination of PPE are under investigation, including use of ultraviolet light and hydrogen peroxide vapor.2 However, sending used respirators to a central processing facility for hydrogen peroxide vapor treatment is likely to be labor-intensive and costly and ultraviolet light is suboptimal for decontamination of soft surfaces.3 There is an urgent need for simple and widely available methods to decontaminate PPE, including cloth masks.
One potential method to decontaminate face masks and respirators is moist or dry heat.3 Previous reports suggest that moist heat at 65°C for 20 minutes is effective against viruses such as influenza.3, 4 In Taiwan, one common practice used for decontamination of cloth face masks is using a short cycle of treatment in a rice cooker or other kitchen steamer. This method has the advantage of being widely available and easy to use, particularly for cloth face masks. Therefore, we evaluated the efficacy of steam treatment applied via a rice cooker-steamer versus similar levels of dry heat for decontamination of cloth and surgical face masks and N95 respirators.
We studied surgical face masks (Precept; Arden, NC), 3M 1860 N95 respirators (3M; Saint Paul, MN), and cotton and quilting fabric cloth face masks being distributed to visitors and personnel not involved in direct patient care at a Cleveland area hospital. The test organisms included a clinical isolate of methicillin-resistant Staphylococcus aureus (MRSA) and the nonenveloped, single-stranded RNA virus bacteriophage MS2 which was prepared as previously described.3 The test protocol has been reported previously.3 In brief, 10-μL aliquots containing 106 colony-forming units (CFU) or plaque-forming units (PFU) of the test organisms suspended in 8% simulated mucus were inoculated onto 1-cm2 areas on both the outer or inner surfaces of the respirators or face masks.3 The inoculated masks or respirators were subjected to a cycle of treatment in a steamer (Aroma; San Diego, CA) lasting approximately 13-15 minutes, including 8-10 minutes of heating and 5 minutes of steam. For comparison, inoculated masks or respirators were subjected to dry heat at 100°C for 15 minutes in an oven (Thermo Fisher Scientific; Waltham, MA). After treatment, the inoculated sections of the face masks and N95 respirators were vortexed for 1 minute in 1 mL of phosphate-buffered saline with 0.02% Tween and serial dilutions were plated on selective media to quantify viable organisms.3 All tests were performed in triplicate. Log10 reductions were calculated in comparison to untreated controls. A reduction of 3-log10 or greater in recovery of organisms inoculated onto masks or respirators was considered effective for decontamination.3
As shown in Figure 1 , the steam treatment resulted in a greater than 5 log10 reduction in bacteriophage MS2 and MRSA applied to the outer and inner surfaces of the face masks and respirators, whereas dry heat at 100°C for 15 minutes did not result in a greater than 3 log10 reduction of either organism at any of the inoculated sites on any masks or respirators. No visible changes were observed in any of the masks or respirators after 5 cycles of decontamination.
Efficacy of treatment in a rice cooker-steamer including 8-10 minutes of heating and 5 minutes of steam versus dry heat at 100°C for 15 minutes in an oven for decontamination of methicillin-resistant Staphylococcus aureus (MRSA) and bacteriophage MS2 on 3M 1860 N95 respirators, surgical face masks, and cotton and quilting fabric cloth face masks. 10-μL aliquots containing 106 colony-forming units (CFU) or plaque-forming units (PFU) of the test organisms in the simulated mucus suspension were spread to cover an area of 1-cm2 on inner and outer surfaces of the N95 respirators and surgical face masks and on one surface of the cloth masks. Error bars indicate standard error.
Our results demonstrate that a short cycle of steam treatment applied via a commonly used kitchen rice cooker-steamer can be very effective for decontamination of face masks and N95 respirators. Dry heat at the same temperature levels was much less effective, consistent with previous evidence that moist heat or microwave-generated steam is more effective than dry heat for inactivation of viruses.4, 5 Notably, the short cycle of steam treatment was substantially more effective than ultraviolet light treatment for N95 decontamination and nearly as effective as aerosolized peracetic acid and hydrogen peroxide.3
The major limitation of our study is that we did not examine the effect of treatment on respirator or face mask performance. However, there is some evidence that short cycles of steam treatment may have minimal effect on filtration and fit performance.2 Further testing is needed to evaluate the impact of steam treatment on performance of N95 respirators and surgical face masks.
In summary, our results demonstrate that steam treatment using a rice cooker-steamer is effective for decontamination of face masks and N95 respirators. Given the recommendation that cloth face masks be worn in public settings, steam treatment using these readily available kitchen items could provide safe and effective decontamination of cloth masks. Further studies are needed to evaluate steam treatment for N95 respirators and surgical face masks. Investigations of moist heat are also needed as 20 minutes of exposure to moist heat at 65°C has been reported to be effective with minimal adverse effects on respirator performance.3, 4 Go to:
This work was supported by the Department of Veterans Affairs.
Potential conflicts of interest. C.J.D has received research grants from Pfizer, Clorox, and PDI. All other authors report no conflicts of interest relevant to this article.Go to:
1. Ranney ML, Griffeth V, Jha AK. Critical Supply Shortages – The Need for Ventilators and Personal Protective Equipment during the Covid-19 Pandemic. N Engl J Med. 2020 Mar 25 doi: 10.1056/NEJMp2006141. [CrossRef] [Google Scholar]2. https://www.cdc.gov/coronavirus/2019-ncov/hcp/ppe-strategy/decontamination-reuse-respirators.html3. Cadnum JL, Li D, Redmond SN, John AR, Pearlmutter B, Donskey CJ. Effectiveness of Ultraviolet-C Light and a High-Level Disinfection Cabinet for Decontamination of N95 Respirators. Pathog Immun. 2020 in press. [Google Scholar]4. Lore MB, Heimbuch BK, Brown TL, Wander JD, Hinrichs SH. Effectiveness of Three Decontamination Treatments against Influenza Virus Applied to Filtering Facepiece Respirators. Ann Occupat Hyg. 2012;56:92–101. doi: 10.1093/annhyg/mer054. [CrossRef] [Google Scholar]5. Heimbuch BK, Wallace WH, Kinney K, Lumley AE, Wu C, Woo M, Wander JD. A pandemic influenza preparedness study: use of energetic methods to decontaminate filtering facepiece respirators contaminated with H1N1 aerosols and droplets. Am J Infect Control. 2011;39:e1–e9. doi: 10.1016/j.ajic.2010.07.004. [PubMed] [CrossRef] [Google Scholar]